CoaguChek® INRange meter
CoaguChek INRange is the connected self-testing meter that enables your patients to test their PT/INR for confident control of their VKA therapy.
Built-in wireless, Bluetooth® technology allows you to be connected to your patients, who can now transmit their PT/INR results from their meter to you. This will allow you to keep patients on track and in range – anytime, anywhere.
CoaguChek technology and quality have been recognised to facilitate VKA patient self-testing in a safe, accurate and reliable manner.1
More time in range
Patients who spend a high proportion of time (> 70%) in the therapeutic range achieve better clinical outcomes.2,3
More engaged patients
Patients who self-test are more engaged in their own care, test more often and spend more time in range.4,5
Value of patient self-testing
Patient self-testing has been shown to result in more time spent in the therapeutic range,6-8 fewer very high or very low INR values,8 fewer thromboembolic events,9-10 improved patient quality of life,11 12 and better treatment satisfaction.11 13 as compared to usual care or anti-coagulation clinic.
When you’re connected, your patients are on track
You’ll be truly connected to your patients, so you can confidently optimise their therapy
The CoaguChek INRange offers more information than just a result in order to encourage a better connection between the patient and healthcare professional. With trending reports, comments and color-coded graphs, you can quickly see where your patients’ PT/INR is heading.
Results trend reporting
The CoaguChek INRange meter will let your patients know if their most recent result is higher or lower than their last result to help understand trends.
Target range aware
Once you setup targets in your INRange, it will clearly let the patient know about results out of that range.
Bluetooth or USB connectivity
Connect the meter using Bluetooth or USB to quickly and easily send results to the healthcare professional. For more information about compatible software for data transfer with the CoaguChek INRange meter please contact your local Roche Diagnostics customer support.
The patient can enter up to 6 comments with each result to help them remember any past actions that may have affected the result.
The color display makes important information such as color coded results noticeable so that they are easy to read and understand.
Reminders and on screen help
Reminders can be setup for important events such as when to test, take medication or next doctor's appointment.
On-screen help provides text guidance through-out the test procedure.
The CoaguChek INRange kit consists of the following components:
- CoaguChek INRange meter
- CoaguChek Softclix lancing device - for individual use only
- 20 CoaguChek Softclix XL lancets
- Carrying case
- User's Manual, including short instruction guide
- USB cable
- 4 x AAA batteries
Additional required components:
- CoaguChek XS PT Test PST strips (available on prescription, or to purchase directly from Roche)
- Getting Started Pack (including step by step guide)
- Training DVD
CoaguChek INRange meter handling video
- Measuring & Sampling
Amperometric (electrochemical) determination of the PT time after activation of the coagulation with human recombinant thromboplastin
Intuitive user interface with color display, on/off, enter, back and up/down button
Outside the meter top- and 2 sides: left or right,-dosing options
- Operating Conditions
+ 15° C to + 32° C (59° F to 90° F)
10 - 85%
4300 m (14,000 ft)
- First option: Operate the monitor on a level surface, free of vibrations
- Second option: Hold the monitor in your hand, so that it is roughly horizontal
Measuring range INR
0.8 - 8.0
USB (Type B) and Bluetooth
Battery - Four 1.5v AAA alkali-manganese batteries
# of tests per set of batteries
Up to 60 tests or 1 year
145 x 75 x 30 mm
135 g (without batteries)
400 Test Results
Auto Power Off
Auto off after 2 minutes of inactivity
- Sample Material
Fresh capillary whole blood
Refer to the test-strip package insert.
- Test Strips
Sensitivity to heparin
Not at therapeutic levels, up to 0.8 I.U/ml for UFH and to 2 anti-Xa U/ml for LMWH
On each strip, through the same channel as the blood passes
21 months from production (+ 2° C to + 30° C)
Display of target range
1. Sharma, P., Scotland, G., Cruickshank, M., Tassie, E., Fraser, C., et al. (2015). Is self-monitoring an effective option for people receiving long-term vitamin K antagonist therapy? A systematic review and economic evaluation. BMJ Open 5:e007758
2. Gallagher, A.M., Setakis, E., Plumb, J.M., Clemens, A., van Staa, T.P. (2011). Risks of stroke and mortality associated with suboptimal anticoagulation in atrial fibrillation patients. Thromb Haemost 106, 968–977
3. Wan, Y., Heneghan, C., Perera, R., Roberts, N., Hollowell, J., et al. (2008). Anticoagulation control and prediction of adverse events in patients with atrial fibrillation: a systematic review. Circ Cardiovasc Qual Outcomes 1, 84–91
4. Ward A, Tompson A, Fitzmaurice D, Sutton S, Perera R., et al. (2015). Cohort study of Anticoagulation Self-Monitoring (CASM): a prospective study of its effectiveness in the community. Br J Gen Pract. 65(636):e428-437
5. Heneghan C., Ward A., Perera R., et al. (2012). Self-monitoring of oral anticoagulation: systematic review and meta-analysis of individual patient data. Lancet 379:322-334
6. Bereznicki, L.R.E., Jackson, S.L., Peterson, G.M. (2013). Supervised patient self-testing of warfarin therapy using an online system. J Med Internet Res 15, e138
7. Christensen, H., Lauterlein, J.-J., Sørensen, P.D., Petersen, E.R.B., Madsen, et al. (2011). Home management of oral anticoagulation via telemedicine versus conventional hospital-based treatment. Telemed J E-Health Off J Am Telemed Assoc 17, 169–176
8. Bussey, H.I., Bussey M., Bussey-Smith K.L., Frei, C.R. (2013). Evaluation of warfarin management with international normalized ratio self-testing and online remote monitoring and management plus low-dose vitamin k with genomic considerations: a pilot study. Pharmacotherapy 33, 1136–1146
9. Heneghan, C., Ward, A., Perera, R., Bankhead, C., Fuller, A., et al. (2012). Self-monitoring of oral anticoagulation: systematic review and meta-analysis of individual patient data. Lancet 379, 322–334
10. Heneghan, C., Alonso-Coello, P., Garcia-Alamino, J.M., Perera, R., Meats, E., Glasziou, P. (2006). Self-monitoring of oral anticoagulation: a systematic review and meta-analysis. Lancet 367, 404–411
11. Matchar, D.B., Jacobson, A., Dolor, R., Edson, R., Uyeda, L., et al. (2010). Effect of home testing of international normalized ratio on clinical events. N Engl J Med 363, 1608–1620
12. Salvador, C.H., Ruiz-Sanchez, A., González de Mingo, M.A., Carmona Rodríguez, M., Carrasco, et al. (2008). Evaluation of a telemedicine based service for the follow-up and monitoring of patients treated with oral anticoagulant therapy. EEE Trans Inf Technol Biomed 12, 696–706
13. Gardiner, C., Williams, K., Mackie, I.J., Machin, S.J., Cohen, H. (2005). Patient self-testing is a reliable and acceptable alternative to laboratory INR monitoring. Br J Haematol 128, 242–247